SETTING THE METER
T h e multimeter must be set to
a voltage range higher than
that being measured or
m u l t i m e t e r could be damaged.
Connect red lead to V-OHMS jack.
Connect black lead to COM jack.
S e t multimeter switches as specified in
Depress appropriate voltage range switch. If
voltage being measured is unknown, select
1000 V range.
Set FUNCTION switch (1) to DIRECT.
Plug black (-) lead (2) in -DC/ħAC/OHMS
To measure 24 volts DC, plug red lead (3) in
5 0 V jack on left side of multimeter. (If
measuring less than 10 volts DC, use 10 V
jack. If measuring less than 2.5 volts DC, use
2.5 V jack.
Plug black lead (1) in COM -jack.
Plug red lead (2) in +jack.
To measure 24 volts DC, set selector switch
(3) to V/DC 50. (If measuring less than 10
volts DC, set selector switch to V/DC 10. If
measuring less than 2.5 V/DC, set selector
switch to V/DC 2.5.)
Set -DC/+DC switch (4) to +DC.