ZEROING THE TS-352 B/U MULTIMETER
For accurate readings, do not
t o u c h probe ends with fingers
while zeroing the multimeter.
Using adjusting screw (1), set needle (2) to
prior to OHMS ZERO ADJ.
Set RANGE switch (3) to RX100.
Plug black (-) lead (4) in OHMS DC ąjack.
Plug red (+) lead (5) in upper left OHMS jack.
Set FUNCTION switch (6) to OHMS.
Hold two probe tips (7) firmly together.
If needle will not zero, replace
batteries in multimeter. If needle
still will not zero, have multimeter
Zero multimeter by turning OHMS ADJ knob
(8) until needle (2) is over 0 on OHMS scale.